My support system ensures that I don’t look totally incompetent. Thank you ben and James , now the machines are working properly, and I can continue my work. Why cannot I Enable Agilent ? The programme works fine and I receive a text data after logging. I changed my VI and now I can configurate the measurements. Put some probes in the initialize VI and see where the error is occurring.
|Date Added:||4 March 2004|
|File Size:||63.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
However, when trying to run the plug-and-play examples, this error occurs “The ID Query failed.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Yes, I have already put probes on the VI.
But I do not Know how I can solve the problem. Is this wrong to do? But I still don’t know how to put my ‘measurement part’ and my ‘logging part’ together.
Agilent Technologies / Keysight Technologies hp34420a Meter
Thank you ben and Jamesnow the machines are working properly, and I can continue my work. Thank you in advance Are your drivers looking for the new scope in the right gailent
My support system ensures that I don’t look totally incompetent. But I’ll try to work on it. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. But it seems as if LabVIEW records the channels step by step and there are always about 5s between two different temperatures or voltages for each channel. Proud to say that I’ve progressed beyond knowing just enough to be dangerous.
Message 5 of Do I really have to use a While Loop or is there a function I don’t know yet? I changed my VI and the reading and scanning works fine. Message 3 of Identification Query Failed Solved! I attached my VI and a text data to show you how it looks like.
Message 6 of But both together is not working the way I want to. You do not understand how to use that instrument correctly. Message 3 of I’d give this a go and see how you get on Like Dennis said look at the examples, my “Scan-now.
A/A LabView Programming Examples | Keysight (formerly Agilent’s Electronic Measurement)
Message 7 of I need to know: Message 9 of The logging part seemed wrong for me too. Message 5 of I did that and still had the same thing.
I am using Agilent A pulse generator for my experiment and trying to use the available driver Agilent Message 6 of The problem is that I don’t know how to choose and change a channel when I don’t split them. The drivers are bugged in my opinion. You may also be using a model that is not officially supported by this driver.